A. Multi-Product Calibrator. Service Manual. PN August Rev.5 , The Fluke Model A Multi-Product Calibrator (Figure ) is a precise. Fluke Corporation. 1. A Specifications. The following paragraphs detail specifications for the A Calibrator. changes more than 5 °C. (See “Zeroing the Calibrator” in Chapter 4 of the A Operator Manual.) Also see additional. 2 Fluke Corporation A/A Multi-Product Calibrators Extended Specifications. Warmup Time. Twice the time since last warmed up, to a maximum of
|Published (Last):||2 May 2009|
|PDF File Size:||1.18 Mb|
|ePub File Size:||16.60 Mb|
|Price:||Free* [*Free Regsitration Required]|
Resistance Accuracy Test cent. To obtain warranty service, contact your nearest Fluke authorized service center or send the product, with a description of the difficulty, postage and insurance prepaid FOB Destinationto the nearest Fluke authorized service center.
Harmonics 2nd to 50th Specifications No Error 1 DDE: Equipment Required The following equipment is necessary for performing the hardware adjustments described in this section. The Calibrator Mainframe must fluke 5500a calibrator manual fully calibrated prior to performing any of the SC calibration procedures.
AC Voltage Verification at 1 Zeroing the Calibrator Got fluke 5500a calibrator manual, continue to print.
Trigger Signal Specifications Edge Fluke 5500a calibrator manual Adjusting the Aberrations for the Edge Function For 2 to 5 mA burden current 33 to Vthe phase uncertainty is 1. AC Volts Measure the A output using a precision ac voltmeter, and enter into the A each manjal the measured values listed in Table when prompted to do so. Once you have turned R44 to the point at which the spur just disappears, the signal is balanced between the VCOs and you have completed the adjustment.
Leveled sine wave amplitude, frequency, harmonics, and flatness Fluke 5500a calibrator manual provided in this manual.
A Operator Manual
DC V oltage Performance Test Apply a short to the Fluke 5500a calibrator manual terminals to provide a low-impedance path for current. Enable the SC30 0 and wait at least 5 minutes.
Harmonics Page – 55500a Sinewave Verification: Fluke 5500a calibrator manual access calibration steps described in paragraphs throughsimply send the command: These shifts are flukw zero unless you have just done a new calibration, but not saved the constants yet. Enter the LCR reading into the A when prompted. Lay the board shield side down.
Elatness calibration of the low frequency band is made relative to 50 kHz. Refer to Eigure to identify the spur. Connect the other end of the BNC cable to the 50 Q.
Thermocouple Measuring This procedure calibrates the temperature measuring capability of the A by externally measuring a known temperature. Table shows the test points. If any provision of this Warranty is held invalid or unenforceable fluke 5500a calibrator manual a court of competent jurisdiction, such holding will not affect the validity or enforceability of any other provision.
See Table and Eigure By pressing ‘print’ button you will print only current page.
A Service Manual
Figure shows the four-wire connections. It will affect the first 50 ns of the wave form. All product names are trademarks of their respective companies. SC Calibration Setup The allowable entry points are as shown in Table Introduction Because this is a high performance instrument, it is not recommended that the user service the hoards to the fluke 5500a calibrator manual fluie.
Thermocouple Measuring Accuracy Test When prompted for the second reference capacitor by the Fluke 5500a calibrator manual Mainframe, connect the 50 pE capacitance to the end of the BNC cable. Record the harmonic level reading for each frequency and harmonic listed in Table Pulse Width Verification The following equipment is used to verify the pulse width.
Capacitance, Four-Wire Fluke 5500a calibrator manual Introduction The Fluke Model A Multi-Produet Calibrator Figure is a preeise instrument that ealibrates a wide variety of eleetrieal measuring instruments. Set the filter on the PM as indicated in the table.
Complete the following procedure to zero the calibrator. You can also source a known temperature from a temperature calibrator using a J-type thermocouple connection and Internal Reference.
Measure the topline first. Press the CAL softkey, opening the calibration information menu. Hardware adjustments must be made to the leveled sine and edge functions each time the SC is repaired. Remove the four Phillips screws that are around the output block.